12/19/23 12:24:36.026	Information	Image Studio Version 5.2.5	GMT -5:00
12/19/23 12:24:36.026	Information	OS: Windows 8 6.2	GMT -5:00
12/19/23 12:24:36.026	Information	Java Home: c:\program files\licor\image studio\jre	GMT -5:00
12/19/23 12:24:36.042	Information	JRE Version: 1.7.0_72 32-bit	GMT -5:00
12/19/23 12:24:36.042	Information	Java library: C:\Program Files\Licor\Image Studio;C:\WINDOWS\Sun\Java\bin;C:\WINDOWS\system32;C:\WINDOWS;C:\Program Files (x86)\Common Files\Oracle\Java\javapath;C:\WINDOWS\system32;C:\WINDOWS;C:\WINDOWS\System32\Wbem;C:\WINDOWS\System32\WindowsPowerShell\v1.0\;C:\WINDOWS\System32\OpenSSH\;C:\Program Files\PolicyPak\;C:\Users\wmao2\AppData\Local\Microsoft\WindowsApps;;c:\program files\licor\image studio\jre\bin;. 32-bit	GMT -5:00
12/19/23 12:24:36.057	Information	Database migrations: 5.0_Initial, 5.0.66_New log entries fks, 5.0.67_Change lane wrappers, 5.2.68_Update database version, 5.2.69_Add Image Id Constraint	GMT -5:00
12/19/23 12:24:36.057	Information	Installed Analysis Key: Grid_Analysis	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Analysis Key: Image_Studio_Base	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Analysis Key: Shapes	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Analysis Key: Western_Analysis	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Instrument Key: CDG_Acquire	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Instrument Key: Odyssey_Acquire	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Instrument Key: OdysseyFc_Acquire	GMT -5:00
12/19/23 12:24:36.073	Information	Installed Instrument Key: OdysseySa_Acquire	GMT -5:00
12/19/23 12:24:36.088	Information	Installed Instrument Key: Pearl_Acquire	GMT -5:00
12/19/23 12:24:36.088	Information	Selected Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/19/23 12:24:36.088	Information	Loading project C:\Users\wmao2\2023-12-19\Images\...	GMT -5:00
12/19/23 12:24:36.151	Information	Loading more acquisitions...	GMT -5:00
12/19/23 12:24:36.526	Information	USBCat Client Version 1.0.2	GMT -5:00
12/19/23 12:24:36.573	Information	Finished adding Acquisitions to Work Area.	GMT -5:00
12/19/23 12:24:36.573	Information	Image Studio Version 5.2.5	GMT -5:00
12/19/23 12:24:43.227	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 12:24:43.227	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 12:24:43.227	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 12:24:43.243	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 12:24:43.290	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 12:24:43.337	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 12:24:43.915	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    640 mV 	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:24:46.305	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1740 mA 	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          30 C  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                31 C  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.93 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.85 V  	GMT -5:00
12/19/23 12:24:46.320	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/19/23 12:24:46.367	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/19/23 12:24:51.335	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          29 C  	GMT -5:00
12/19/23 12:24:55.350	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          28 C  	GMT -5:00
12/19/23 12:24:59.364	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          27 C  	GMT -5:00
12/19/23 12:25:04.379	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          26 C  	GMT -5:00
12/19/23 12:25:08.394	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1710 mA 	GMT -5:00
12/19/23 12:25:08.456	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:25:10.409	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1669 mA 	GMT -5:00
12/19/23 12:25:11.409	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1628 mA 	GMT -5:00
12/19/23 12:25:12.408	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1577 mA 	GMT -5:00
12/19/23 12:25:13.408	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1516 mA 	GMT -5:00
12/19/23 12:25:13.471	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          24 C  	GMT -5:00
12/19/23 12:25:14.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1445 mA 	GMT -5:00
12/19/23 12:25:15.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1354 mA 	GMT -5:00
12/19/23 12:25:16.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1252 mA 	GMT -5:00
12/19/23 12:25:17.423	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1150 mA 	GMT -5:00
12/19/23 12:25:18.438	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                     1038 mA 	GMT -5:00
12/19/23 12:25:18.485	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          23 C  	GMT -5:00
12/19/23 12:25:19.313	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 12:25:19.438	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      936 mA 	GMT -5:00
12/19/23 12:25:19.485	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.91 V  	GMT -5:00
12/19/23 12:25:20.438	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      835 mA 	GMT -5:00
12/19/23 12:25:21.453	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      733 mA 	GMT -5:00
12/19/23 12:25:22.453	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      641 mA 	GMT -5:00
12/19/23 12:25:23.453	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      560 mA 	GMT -5:00
12/19/23 12:25:24.452	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      478 mA 	GMT -5:00
12/19/23 12:25:25.468	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      407 mA 	GMT -5:00
12/19/23 12:25:26.468	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      346 mA 	GMT -5:00
12/19/23 12:25:27.467	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/19/23 12:25:27.530	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/19/23 12:25:28.233	Information	[Instrument Request]  1 ScanAutoEstimate 169um lowest 0.0 0.1 5.0 0.0 5.7 700 800 	GMT -5:00
12/19/23 12:25:28.248	Information	[Instrument Response] 1 Status 0 Executing command: ScanAutoEstimate 169um lowest 0 0.1 5 0 5.7 700 800	GMT -5:00
12/19/23 12:25:28.295	Information	[Instrument Response] 1 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:25:28.295	Information	[Instrument Response] 1 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:25:28.311	Information	[Instrument Response] 1 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:25:28.311	Information	[Instrument Response] 1 Status 0 Y mapping: [y0=0,y1=5.7] => [top=34,bottom=2730] h=337	GMT -5:00
12/19/23 12:25:28.327	Information	[Instrument Response] 1 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:25:28.327	Information	[Instrument Response] 1 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:25:28.327	Information	[Instrument Response] 1 Status 0 Seconds/line=0.229049	GMT -5:00
12/19/23 12:25:28.327	Information	[Instrument Response] 1 Status 0 y_scan_pixels=337, passes=2 => lines_to_scan=674	GMT -5:00
12/19/23 12:25:28.342	Information	[Instrument Response] 1 Status 0 linesPerBlock=44 flushLinesPerBlock=22	GMT -5:00
12/19/23 12:25:28.342	Information	[Instrument Response] 1 Status 0 Blocks=16	GMT -5:00
12/19/23 12:25:28.358	Information	[Instrument Response] 1 Status 0 Estimates (sec): setup=2.02 line=0.242792 => total=166	GMT -5:00
12/19/23 12:25:28.358	Information	[Instrument Response] 1 Progress 0% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:25:28.358	Information	[Instrument Response] 1 Result 0 OK	GMT -5:00
12/19/23 12:25:28.467	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/19/23 12:25:28.530	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          22 C  	GMT -5:00
12/19/23 12:25:29.483	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      204 mA 	GMT -5:00
12/19/23 12:25:30.154	Information	[Instrument Request]  2 ScanAutoEstimate 169um lowest 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:25:30.170	Information	[Instrument Response] 2 Status 0 Executing command: ScanAutoEstimate 169um lowest 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:25:30.217	Information	[Instrument Response] 2 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:25:30.217	Information	[Instrument Response] 2 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:25:30.217	Information	[Instrument Response] 2 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:25:30.232	Information	[Instrument Response] 2 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:25:30.232	Information	[Instrument Response] 2 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:25:30.248	Information	[Instrument Response] 2 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:25:30.248	Information	[Instrument Response] 2 Status 0 Seconds/line=0.229049	GMT -5:00
12/19/23 12:25:30.264	Information	[Instrument Response] 2 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:25:30.264	Information	[Instrument Response] 2 Status 0 linesPerBlock=44 flushLinesPerBlock=22	GMT -5:00
12/19/23 12:25:30.264	Information	[Instrument Response] 2 Status 0 Blocks=17	GMT -5:00
12/19/23 12:25:30.264	Information	[Instrument Response] 2 Status 0 Estimates (sec): setup=2.02 line=0.242792 => total=175	GMT -5:00
12/19/23 12:25:30.279	Information	[Instrument Response] 2 Progress 0% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:25:30.279	Information	[Instrument Response] 2 Result 0 OK	GMT -5:00
12/19/23 12:25:30.482	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      163 mA 	GMT -5:00
12/19/23 12:25:31.482	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      132 mA 	GMT -5:00
12/19/23 12:25:31.998	Information	[Instrument Request]  3 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:25:32.013	Information	[Instrument Response] 3 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Blocks=25	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:25:32.044	Information	[Instrument Response] 3 Result 0 OK	GMT -5:00
12/19/23 12:25:32.482	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      102 mA 	GMT -5:00
12/19/23 12:25:34.497	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       61 mA 	GMT -5:00
12/19/23 12:25:35.341	Information	Starting acquisition...	GMT -5:00
12/19/23 12:25:36.215	Information	[Instrument Request]  4 ScanAuto 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:25:36.309	Information	[Instrument Response] 4 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Blocks=25	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:25:36.325	Information	[Instrument Response] 4 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:25:36.340	Information	[Instrument Response] 4 Progress 0% Initializing	GMT -5:00
12/19/23 12:25:36.340	Information	[Instrument Response] 4 Status 0 Preparation complete	GMT -5:00
12/19/23 12:25:36.497	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       31 mA 	GMT -5:00
12/19/23 12:25:39.230	Information	[Instrument Response] 4 Status 0 Motors initialized	GMT -5:00
12/19/23 12:25:39.293	Information	[Instrument Response] 4 Status 0 X-Motor configured	GMT -5:00
12/19/23 12:25:39.293	Information	[Instrument Response] 4 Status 0 DSP configured	GMT -5:00
12/19/23 12:25:39.340	Information	[Instrument Response] 4 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 12:25:39.340	Information	[Instrument Response] 4 Status 0 Cooling	GMT -5:00
12/19/23 12:25:39.340	Information	[Instrument Response] 4 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 12:25:39.512	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 12:25:46.541	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          23 C  	GMT -5:00
12/19/23 12:26:02.600	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          24 C  	GMT -5:00
12/19/23 12:26:10.645	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       61 mA 	GMT -5:00
12/19/23 12:26:15.660	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                       92 mA 	GMT -5:00
12/19/23 12:26:19.674	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      122 mA 	GMT -5:00
12/19/23 12:26:24.689	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      153 mA 	GMT -5:00
12/19/23 12:26:25.704	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:26:31.718	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      183 mA 	GMT -5:00
12/19/23 12:26:37.748	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:26:38.295	Information	[Instrument Response] 4 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 355 xofs=1 yofs=0	GMT -5:00
12/19/23 12:26:38.342	Information	[Instrument Response] 4 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 355 xofs=0 yofs=0	GMT -5:00
12/19/23 12:26:38.342	Information	[Instrument Response] 4 Status 0 start=34 scan_lines=15 line=0 end=355 700=Low,L1 800=High,4	GMT -5:00
12/19/23 12:26:38.342	Information	[Instrument Response] 4 Status 0 start=34 scan_lines=30 line=0 end=355 700=High,4 800=Low,L1	GMT -5:00
12/19/23 12:26:38.342	Information	[Instrument Response] 4 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 12:26:38.748	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:26:39.748	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:26:42.357	Information	[Instrument Response] 4 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:26:42.763	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:26:42.825	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:26:42.825	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:26:42.825	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:26:42.825	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:26:44.778	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      214 mA 	GMT -5:00
12/19/23 12:26:46.777	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:26:47.777	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/19/23 12:26:50.792	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:26:50.855	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/19/23 12:26:50.855	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:26:50.855	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:26:51.808	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:26:56.822	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/19/23 12:26:57.634	Information	[Instrument Response] 4 ImageBlock 700 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:26:57.634	Information	[Instrument Response] 4 ImageBlock 800 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:26:57.650	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:26:57.697	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:27:01.837	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:27:01.915	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:27:02.852	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:02.899	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:27:02.977	Information	[Instrument Response] 4 Progress 6% 5 minutes	GMT -5:00
12/19/23 12:27:03.852	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/19/23 12:27:03.899	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:27:03.899	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:27:03.899	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:27:04.851	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:27:09.866	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:27:10.881	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:10.991	Information	[Instrument Response] 4 ImageBlock 700 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:10.991	Information	[Instrument Response] 4 ImageBlock 800 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:11.006	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:27:11.038	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:27:16.911	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:27:16.974	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         136 V  	GMT -5:00
12/19/23 12:27:16.974	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:27:16.974	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:27:16.974	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:27:17.911	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:27:17.973	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:18.911	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:27:24.363	Information	[Instrument Response] 4 ImageBlock 700 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:24.363	Information	[Instrument Response] 4 ImageBlock 800 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:24.363	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:27:24.394	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:27:30.971	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:27:31.017	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:27:31.017	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:27:31.017	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:27:33.220	Information	[Instrument Response] 4 Progress 15% 4 minutes, 30 seconds	GMT -5:00
12/19/23 12:27:33.970	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:27:34.985	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:36.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:27:37.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:27:37.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:27:37.719	Information	[Instrument Response] 4 ImageBlock 700 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:37.719	Information	[Instrument Response] 4 ImageBlock 800 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:37.719	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:27:37.766	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:27:37.985	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:27:38.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:38.047	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:27:44.014	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:27:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/19/23 12:27:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:27:44.077	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:27:45.014	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:27:51.091	Information	[Instrument Response] 4 ImageBlock 700 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:51.091	Information	[Instrument Response] 4 ImageBlock 800 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:27:51.091	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:27:51.122	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:27:57.074	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 12:27:57.121	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:27:57.121	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:27:57.121	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:27:57.121	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:27:58.074	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:27:58.121	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:27:59.074	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:28:01.089	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:28:02.088	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:28:03.354	Information	[Instrument Response] 4 Progress 25% 4 minutes	GMT -5:00
12/19/23 12:28:04.088	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:28:04.447	Information	[Instrument Response] 4 ImageBlock 700 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:04.447	Information	[Instrument Response] 4 ImageBlock 800 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:04.447	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:28:04.479	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:28:05.103	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:28:11.133	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:28:11.180	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:28:11.180	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:28:11.180	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:28:12.133	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:28:17.819	Information	[Instrument Response] 4 ImageBlock 700 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:17.819	Information	[Instrument Response] 4 ImageBlock 800 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:17.819	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:28:17.851	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:28:24.177	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:28:24.240	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:28:24.240	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:28:24.240	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:28:25.177	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:28:31.176	Information	[Instrument Response] 4 ImageBlock 700 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:31.176	Information	[Instrument Response] 4 ImageBlock 800 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:31.176	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:28:31.222	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:28:31.285	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:28:31.285	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:28:32.207	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:28:32.269	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:28:33.503	Information	[Instrument Response] 4 Progress 35% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:28:34.222	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          56 mA 	GMT -5:00
12/19/23 12:28:34.269	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:28:35.222	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:28:35.284	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:28:37.237	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:28:37.284	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/19/23 12:28:37.284	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:28:37.284	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:28:38.237	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:28:39.236	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:28:44.548	Information	[Instrument Response] 4 ImageBlock 700 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:44.548	Information	[Instrument Response] 4 ImageBlock 800 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:44.548	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:28:44.579	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:28:50.281	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         150 V  	GMT -5:00
12/19/23 12:28:50.343	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         158 V  	GMT -5:00
12/19/23 12:28:50.343	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:28:50.343	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:28:50.343	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:28:51.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/19/23 12:28:51.343	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:28:51.343	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:28:52.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:28:57.904	Information	[Instrument Response] 4 ImageBlock 700 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:57.904	Information	[Instrument Response] 4 ImageBlock 800 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:28:57.920	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:28:57.935	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:28:58.326	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:28:58.373	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:28:59.325	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:28:59.372	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:29:01.325	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:29:02.340	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:29:03.246	Information	[Instrument Response] 4 Progress 44% 3 minutes	GMT -5:00
12/19/23 12:29:04.340	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:29:04.402	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         103 V  	GMT -5:00
12/19/23 12:29:04.402	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:29:04.402	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:29:05.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:29:11.276	Information	[Instrument Response] 4 ImageBlock 700 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:11.276	Information	[Instrument Response] 4 ImageBlock 800 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:11.276	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:29:11.307	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:29:17.401	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         118 V  	GMT -5:00
12/19/23 12:29:17.479	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:29:17.479	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:29:17.479	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:29:18.401	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:29:24.634	Information	[Instrument Response] 4 ImageBlock 700 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:24.634	Information	[Instrument Response] 4 ImageBlock 800 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:24.649	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:29:24.665	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:29:25.430	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:29:26.446	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:29:30.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:29:30.507	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         144 V  	GMT -5:00
12/19/23 12:29:30.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:29:30.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:29:30.507	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:29:31.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/19/23 12:29:31.507	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:29:32.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:29:33.475	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:29:33.522	Information	[Instrument Response] 4 Progress 53% 2 minutes, 30 seconds	GMT -5:00
12/19/23 12:29:34.475	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:29:38.005	Information	[Instrument Response] 4 ImageBlock 700 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:38.005	Information	[Instrument Response] 4 ImageBlock 800 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:38.005	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:29:38.037	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:29:44.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          97 V  	GMT -5:00
12/19/23 12:29:44.575	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:29:44.575	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:29:44.575	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:29:45.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:29:51.370	Information	[Instrument Response] 4 ImageBlock 700 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:51.370	Information	[Instrument Response] 4 ImageBlock 800 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:29:51.370	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:29:51.402	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:29:57.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:29:57.650	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/19/23 12:29:57.650	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:29:57.650	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:29:58.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:30:00.587	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:30:00.650	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:30:01.587	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:30:01.649	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:30:03.258	Information	[Instrument Response] 4 Progress 63% 2 minutes	GMT -5:00
12/19/23 12:30:04.727	Information	[Instrument Response] 4 ImageBlock 700 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:04.727	Information	[Instrument Response] 4 ImageBlock 800 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:04.742	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:30:04.758	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:30:10.632	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         136 V  	GMT -5:00
12/19/23 12:30:10.694	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:30:10.694	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:30:10.694	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:30:10.694	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:30:11.631	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:30:11.694	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:30:12.647	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:30:13.600	Information	[Instrument Response] 4 Progress 66% 1 minute, 50 seconds	GMT -5:00
12/19/23 12:30:18.099	Information	[Instrument Response] 4 ImageBlock 700 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:18.099	Information	[Instrument Response] 4 ImageBlock 800 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:18.099	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:30:18.130	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:30:22.676	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:30:22.738	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:30:23.441	Information	[Instrument Response] 4 Progress 70% 1 minute, 40 seconds	GMT -5:00
12/19/23 12:30:23.691	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:30:23.738	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:30:24.691	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:30:24.753	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:30:24.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:30:24.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:30:27.706	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:30:30.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:30:31.455	Information	[Instrument Response] 4 ImageBlock 700 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:31.471	Information	[Instrument Response] 4 ImageBlock 800 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:31.471	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:30:31.486	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:30:31.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:30:32.720	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:30:33.392	Information	[Instrument Response] 4 Progress 73% 1 minute, 30 seconds	GMT -5:00
12/19/23 12:30:37.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/19/23 12:30:37.813	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:30:37.813	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:30:37.813	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:30:38.766	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:30:41.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:30:43.359	Information	[Instrument Response] 4 Progress 75% 1 minute, 20 seconds	GMT -5:00
12/19/23 12:30:44.827	Information	[Instrument Response] 4 ImageBlock 700 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:44.827	Information	[Instrument Response] 4 ImageBlock 800 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:44.827	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:30:44.858	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:30:50.810	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:30:50.857	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         129 V  	GMT -5:00
12/19/23 12:30:50.857	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:30:50.857	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:30:51.810	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:30:52.810	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:30:53.310	Information	[Instrument Response] 4 Progress 78% 1 minute, 10 seconds	GMT -5:00
12/19/23 12:30:57.840	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:30:57.887	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:30:58.184	Information	[Instrument Response] 4 ImageBlock 700 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:58.184	Information	[Instrument Response] 4 ImageBlock 800 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:30:58.199	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:30:58.215	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:30:58.840	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:30:58.887	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:31:03.526	Information	[Instrument Response] 4 Progress 82% 1 minute	GMT -5:00
12/19/23 12:31:04.870	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:31:04.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:31:04.917	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:31:04.917	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:31:05.870	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:31:11.556	Information	[Instrument Response] 4 ImageBlock 700 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:11.556	Information	[Instrument Response] 4 ImageBlock 800 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:11.556	Information	Transferring image block at (0,285) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:31:11.571	Information	Transferring image block at (0,285) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:31:13.493	Information	[Instrument Response] 4 Progress 85% 50 seconds	GMT -5:00
12/19/23 12:31:16.914	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:31:17.914	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:31:17.992	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/19/23 12:31:17.992	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:31:17.992	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:31:18.929	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:31:23.444	Information	[Instrument Response] 4 Progress 88% 40 seconds	GMT -5:00
12/19/23 12:31:24.913	Information	[Instrument Response] 4 ImageBlock 700 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:24.913	Information	[Instrument Response] 4 ImageBlock 800 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:24.928	Information	Transferring image block at (0,300) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:31:24.944	Information	Transferring image block at (0,300) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:31:24.991	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    653 mV 	GMT -5:00
12/19/23 12:31:25.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:31:25.006	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:31:25.944	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:31:26.006	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:31:30.974	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         123 V  	GMT -5:00
12/19/23 12:31:31.036	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:31:31.036	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:31:31.036	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:31:31.974	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:31:33.395	Information	[Instrument Response] 4 Progress 91% 30 seconds	GMT -5:00
12/19/23 12:31:38.285	Information	[Instrument Response] 4 ImageBlock 700 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:38.285	Information	[Instrument Response] 4 ImageBlock 800 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:38.285	Information	Transferring image block at (0,315) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:31:38.316	Information	Transferring image block at (0,315) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:31:43.628	Information	[Instrument Response] 4 Progress 95% 20 seconds	GMT -5:00
12/19/23 12:31:44.034	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         156 V  	GMT -5:00
12/19/23 12:31:44.081	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         151 V  	GMT -5:00
12/19/23 12:31:44.081	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:31:44.081	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          35 mA 	GMT -5:00
12/19/23 12:31:45.033	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:31:45.080	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/19/23 12:31:46.033	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:31:51.657	Information	[Instrument Response] 4 ImageBlock 700 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:51.657	Information	[Instrument Response] 4 ImageBlock 800 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:31:51.657	Information	Transferring image block at (0,330) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:31:51.673	Information	Transferring image block at (0,330) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:31:52.063	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:31:52.110	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:31:53.063	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/19/23 12:31:53.126	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:31:53.579	Information	[Instrument Response] 4 Progress 98% 10 seconds	GMT -5:00
12/19/23 12:31:55.078	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:31:56.078	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         107 V  	GMT -5:00
12/19/23 12:31:56.125	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:31:56.125	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:31:56.125	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:31:56.125	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:31:56.125	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:31:57.078	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:32:01.265	Information	[Instrument Response] 4 Progress 100% 0 seconds	GMT -5:00
12/19/23 12:32:02.108	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:32:02.155	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:32:02.155	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:32:02.155	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:32:03.264	Information	[Instrument Response] 4 ImageBlock 700 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:32:03.264	Information	[Instrument Response] 4 ImageBlock 800 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:32:03.264	Information	Transferring image block at (0,345) with size 290x10 to the 700 channel.	GMT -5:00
12/19/23 12:32:03.264	Information	[Instrument Response] 4 Status 0 COMPLETE SCAN 320.902 seconds	GMT -5:00
12/19/23 12:32:03.295	Information	Transferring image block at (0,345) with size 290x10 to the 800 channel.	GMT -5:00
12/19/23 12:32:03.295	Information	[Instrument Response] 4 Status 0 Averages (sec): setup=2.00005 line=0.378733 => total=271	GMT -5:00
12/19/23 12:32:03.342	Information	[Instrument Response] 4 Status 0 Comparisons: setup=1.00997 line=0.939768	GMT -5:00
12/19/23 12:32:03.342	Information	[Instrument Response] 4 Status 0 Finalize motors	GMT -5:00
12/19/23 12:32:03.342	Information	[Instrument Response] 4 Progress 100% Scan complete	GMT -5:00
12/19/23 12:32:03.389	Information	[Instrument Response] 4 Result 0 OK	GMT -5:00
12/19/23 12:32:03.389	Information	Acquisition 0000027_01 finished with channels: 700 800.	GMT -5:00
12/19/23 12:32:03.858	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 12:32:03.858	Information	Scan completed successfully	GMT -5:00
12/19/23 12:32:04.108	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          97 V  	GMT -5:00
12/19/23 12:32:04.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          61 V  	GMT -5:00
12/19/23 12:32:04.170	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:32:04.170	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:32:04.170	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:32:05.116	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          59 V  	GMT -5:00
12/19/23 12:32:05.169	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          38 V  	GMT -5:00
12/19/23 12:32:06.123	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          36 V  	GMT -5:00
12/19/23 12:32:06.178	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          23 V  	GMT -5:00
12/19/23 12:32:07.129	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          24 V  	GMT -5:00
12/19/23 12:32:07.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          15 V  	GMT -5:00
12/19/23 12:32:08.130	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          17 V  	GMT -5:00
12/19/23 12:32:08.179	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          10 V  	GMT -5:00
12/19/23 12:32:08.179	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/19/23 12:32:09.131	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          11 V  	GMT -5:00
12/19/23 12:32:09.189	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/19/23 12:32:10.141	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           8 V  	GMT -5:00
12/19/23 12:32:10.200	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/19/23 12:32:11.135	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/19/23 12:32:12.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/19/23 12:32:13.163	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/19/23 12:32:16.179	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:32:17.193	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:33:25.440	Information	Successfully saved C:\Users\wmao2\2023-12-19\Inhibin A.tif	GMT -5:00
12/19/23 12:33:28.674	Information	[Instrument Request]  5 Stop 	GMT -5:00
12/19/23 12:33:28.721	Information	[Instrument Response] 5 Result 0 OK	GMT -5:00
12/19/23 12:33:29.439	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
12/19/23 12:34:12.181	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 12:34:12.181	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 12:34:12.181	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 12:34:12.181	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 12:34:12.259	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 12:34:12.290	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 12:34:12.306	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 12:34:12.306	Information	[Instrument Request]  6 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Blocks=25	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 6 Result 0 OK	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    653 mV 	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:34:15.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.98 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/19/23 12:34:15.274	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/19/23 12:34:15.321	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 12:34:17.836	Information	Starting acquisition...	GMT -5:00
12/19/23 12:34:18.648	Information	[Instrument Request]  7 ScanAuto 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:34:18.695	Information	[Instrument Response] 7 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Blocks=25	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:34:18.773	Information	[Instrument Response] 7 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:34:18.789	Information	[Instrument Response] 7 Progress 0% Initializing	GMT -5:00
12/19/23 12:34:18.804	Information	[Instrument Response] 7 Status 0 Preparation complete	GMT -5:00
12/19/23 12:34:21.663	Information	[Instrument Response] 7 Status 0 Motors initialized	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 X-Motor configured	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 DSP configured	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Cooling	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 355 xofs=1 yofs=0	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 355 xofs=0 yofs=0	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 start=34 scan_lines=15 line=0 end=355 700=Low,L1 800=High,4	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 start=34 scan_lines=30 line=0 end=355 700=High,4 800=Low,L1	GMT -5:00
12/19/23 12:34:21.726	Information	[Instrument Response] 7 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 12:34:22.288	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 12:34:22.351	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:34:25.303	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:34:26.397	Information	[Instrument Response] 7 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:34:27.303	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:34:27.365	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:34:27.365	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:34:27.365	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:34:27.365	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:34:32.333	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/19/23 12:34:34.332	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:34:34.395	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         129 V  	GMT -5:00
12/19/23 12:34:34.395	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:34:34.395	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:34:34.395	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:34:35.348	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:34:35.395	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:34:36.348	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:34:41.706	Information	[Instrument Response] 7 ImageBlock 700 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:34:41.706	Information	[Instrument Response] 7 ImageBlock 800 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:34:41.706	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:34:41.737	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:34:43.377	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/19/23 12:34:44.377	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:34:45.377	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:34:47.439	Information	[Instrument Response] 7 Progress 6% 5 minutes	GMT -5:00
12/19/23 12:34:48.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:34:48.455	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:34:48.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:34:48.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:34:49.407	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:34:55.078	Information	[Instrument Response] 7 ImageBlock 700 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:34:55.078	Information	[Instrument Response] 7 ImageBlock 800 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:34:55.078	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:34:55.109	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:35:01.452	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:35:01.514	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:35:01.514	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:35:01.514	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:35:02.452	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:35:08.435	Information	[Instrument Response] 7 ImageBlock 700 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:08.435	Information	[Instrument Response] 7 ImageBlock 800 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:08.450	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:35:08.466	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:35:08.529	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:35:09.482	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:35:11.497	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:35:11.544	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:35:12.497	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:35:12.559	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:35:14.512	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         121 V  	GMT -5:00
12/19/23 12:35:14.559	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:35:14.559	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:35:14.559	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:35:15.512	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:35:16.511	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:35:17.511	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:35:17.683	Information	[Instrument Response] 7 Progress 15% 4 minutes, 30 seconds	GMT -5:00
12/19/23 12:35:21.807	Information	[Instrument Response] 7 ImageBlock 700 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:21.807	Information	[Instrument Response] 7 ImageBlock 800 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:21.807	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:35:21.838	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:35:27.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/19/23 12:35:27.619	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         149 V  	GMT -5:00
12/19/23 12:35:27.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:35:27.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:35:27.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:35:27.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:35:28.571	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:35:28.634	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          87 V  	GMT -5:00
12/19/23 12:35:28.634	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:35:28.634	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:35:29.571	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:35:35.164	Information	[Instrument Response] 7 ImageBlock 700 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:35.164	Information	[Instrument Response] 7 ImageBlock 800 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:35.179	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:35:35.195	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:35:38.616	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:35:38.663	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:35:39.616	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:35:39.679	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:35:41.631	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         100 V  	GMT -5:00
12/19/23 12:35:41.678	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:35:41.678	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:35:41.678	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:35:42.631	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:35:47.443	Information	[Instrument Response] 7 Progress 25% 4 minutes	GMT -5:00
12/19/23 12:35:48.536	Information	[Instrument Response] 7 ImageBlock 700 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:48.536	Information	[Instrument Response] 7 ImageBlock 800 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:35:48.536	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:35:48.567	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:35:54.691	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:35:54.738	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         118 V  	GMT -5:00
12/19/23 12:35:54.738	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:35:54.738	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:35:55.691	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:36:01.909	Information	[Instrument Response] 7 ImageBlock 700 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:01.909	Information	[Instrument Response] 7 ImageBlock 800 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:01.909	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:36:01.924	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:36:05.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:36:05.783	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:36:06.736	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:36:06.783	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:36:07.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         141 V  	GMT -5:00
12/19/23 12:36:07.798	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:36:07.798	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:36:07.798	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:36:07.798	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:36:07.798	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:36:08.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          86 V  	GMT -5:00
12/19/23 12:36:08.798	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:36:08.798	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:36:08.798	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:36:09.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:36:13.766	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:36:14.766	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:36:15.266	Information	[Instrument Response] 7 ImageBlock 700 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:15.266	Information	[Instrument Response] 7 ImageBlock 800 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:15.266	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:36:15.297	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:36:16.781	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          55 mA 	GMT -5:00
12/19/23 12:36:17.578	Information	[Instrument Response] 7 Progress 35% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:36:17.796	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:36:21.811	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:36:21.858	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          97 V  	GMT -5:00
12/19/23 12:36:21.858	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:36:21.858	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:36:28.638	Information	[Instrument Response] 7 ImageBlock 700 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:28.638	Information	[Instrument Response] 7 ImageBlock 800 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:28.638	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:36:28.654	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:36:34.855	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/19/23 12:36:34.918	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:36:34.918	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:36:34.918	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:36:35.855	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:36:37.870	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:36:38.870	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:36:40.885	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:36:40.932	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:36:41.885	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:36:41.995	Information	[Instrument Response] 7 ImageBlock 700 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:41.995	Information	[Instrument Response] 7 ImageBlock 800 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:41.995	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:36:42.026	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:36:47.728	Information	[Instrument Response] 7 Progress 44% 3 minutes	GMT -5:00
12/19/23 12:36:47.915	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:36:47.978	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         135 V  	GMT -5:00
12/19/23 12:36:47.978	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:36:47.978	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:36:47.978	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:36:48.915	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:36:48.978	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:36:49.930	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:36:54.945	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:36:55.367	Information	[Instrument Response] 7 ImageBlock 700 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:55.367	Information	[Instrument Response] 7 ImageBlock 800 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:36:55.367	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:36:55.398	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:37:01.975	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:37:02.022	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:37:02.022	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:37:02.022	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:37:02.975	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:37:07.989	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:37:08.724	Information	[Instrument Response] 7 ImageBlock 700 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:08.724	Information	[Instrument Response] 7 ImageBlock 800 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:08.724	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:37:08.755	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:37:09.005	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:37:11.004	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:37:12.020	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:37:15.035	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:37:15.097	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         108 V  	GMT -5:00
12/19/23 12:37:15.097	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:37:15.097	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:37:16.035	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:37:17.581	Information	[Instrument Response] 7 Progress 53% 2 minutes, 30 seconds	GMT -5:00
12/19/23 12:37:22.096	Information	[Instrument Response] 7 ImageBlock 700 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:22.096	Information	[Instrument Response] 7 ImageBlock 800 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:22.096	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:37:22.127	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:37:28.095	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/19/23 12:37:28.157	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:37:28.157	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:37:28.157	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:37:28.157	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:37:28.157	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:37:29.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:37:29.157	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:37:29.157	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:37:30.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:37:35.452	Information	[Instrument Response] 7 ImageBlock 700 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:35.452	Information	[Instrument Response] 7 ImageBlock 800 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:35.468	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:37:35.484	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:37:38.139	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:37:40.155	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:37:41.154	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:37:42.154	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:37:42.217	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:37:42.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:37:42.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:37:42.217	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:37:43.154	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:37:47.731	Information	[Instrument Response] 7 Progress 63% 2 minutes	GMT -5:00
12/19/23 12:37:48.825	Information	[Instrument Response] 7 ImageBlock 700 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:48.825	Information	[Instrument Response] 7 ImageBlock 800 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:37:48.825	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:37:48.856	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:37:55.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/19/23 12:37:55.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:37:55.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:37:55.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:37:56.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:37:57.682	Information	[Instrument Response] 7 Progress 66% 1 minute, 50 seconds	GMT -5:00
12/19/23 12:38:02.181	Information	[Instrument Response] 7 ImageBlock 700 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:02.181	Information	[Instrument Response] 7 ImageBlock 800 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:02.197	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:38:02.213	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:38:05.259	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:38:05.306	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:38:06.259	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:38:07.524	Information	[Instrument Response] 7 Progress 70% 1 minute, 40 seconds	GMT -5:00
12/19/23 12:38:08.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:38:08.321	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:38:08.321	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/19/23 12:38:08.321	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:38:08.321	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:38:09.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:38:10.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:38:11.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:38:13.288	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:38:14.288	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:38:15.554	Information	[Instrument Response] 7 ImageBlock 700 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:15.554	Information	[Instrument Response] 7 ImageBlock 800 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:15.554	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:38:15.585	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:38:17.491	Information	[Instrument Response] 7 Progress 73% 1 minute, 30 seconds	GMT -5:00
12/19/23 12:38:21.334	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         147 V  	GMT -5:00
12/19/23 12:38:21.381	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:38:21.381	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:38:21.381	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:38:21.381	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:38:21.381	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:38:22.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          86 V  	GMT -5:00
12/19/23 12:38:22.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:38:22.396	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:38:23.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:38:27.442	Information	[Instrument Response] 7 Progress 75% 1 minute, 20 seconds	GMT -5:00
12/19/23 12:38:28.910	Information	[Instrument Response] 7 ImageBlock 700 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:28.926	Information	[Instrument Response] 7 ImageBlock 800 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:28.926	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:38:28.942	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:38:30.363	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:38:31.379	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:38:35.393	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:38:35.440	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         100 V  	GMT -5:00
12/19/23 12:38:35.440	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:38:35.440	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:38:36.393	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:38:37.393	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:38:37.455	Information	[Instrument Response] 7 Progress 78% 1 minute, 10 seconds	GMT -5:00
12/19/23 12:38:40.408	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:38:41.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:38:41.470	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:38:41.470	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:38:42.283	Information	[Instrument Response] 7 ImageBlock 700 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:42.283	Information	[Instrument Response] 7 ImageBlock 800 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:42.283	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:38:42.314	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:38:47.625	Information	[Instrument Response] 7 Progress 82% 1 minute	GMT -5:00
12/19/23 12:38:48.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         116 V  	GMT -5:00
12/19/23 12:38:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:38:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:38:48.500	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:38:49.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:38:55.655	Information	[Instrument Response] 7 ImageBlock 700 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:55.655	Information	[Instrument Response] 7 ImageBlock 800 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:38:55.655	Information	Transferring image block at (0,285) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:38:55.670	Information	Transferring image block at (0,285) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:38:57.592	Information	[Instrument Response] 7 Progress 85% 50 seconds	GMT -5:00
12/19/23 12:39:01.497	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:39:01.560	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         140 V  	GMT -5:00
12/19/23 12:39:01.560	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:39:01.560	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:39:01.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:39:01.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:39:02.513	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/19/23 12:39:02.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:39:03.513	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:39:04.512	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:39:07.527	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:39:07.574	Information	[Instrument Response] 7 Progress 88% 40 seconds	GMT -5:00
12/19/23 12:39:08.527	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    710 mV 	GMT -5:00
12/19/23 12:39:09.011	Information	[Instrument Response] 7 ImageBlock 700 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:09.011	Information	[Instrument Response] 7 ImageBlock 800 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:09.011	Information	Transferring image block at (0,300) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:39:09.043	Information	Transferring image block at (0,300) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:39:10.542	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:39:11.542	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:39:15.573	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:39:15.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:39:15.620	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:39:15.620	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:39:17.494	Information	[Instrument Response] 7 Progress 91% 30 seconds	GMT -5:00
12/19/23 12:39:22.384	Information	[Instrument Response] 7 ImageBlock 700 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:22.384	Information	[Instrument Response] 7 ImageBlock 800 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:22.384	Information	Transferring image block at (0,315) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:39:22.399	Information	Transferring image block at (0,315) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:39:27.726	Information	[Instrument Response] 7 Progress 95% 20 seconds	GMT -5:00
12/19/23 12:39:28.632	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:39:28.695	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/19/23 12:39:28.695	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:39:28.695	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:39:29.632	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:39:35.740	Information	[Instrument Response] 7 ImageBlock 700 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:35.740	Information	[Instrument Response] 7 ImageBlock 800 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:39:35.740	Information	Transferring image block at (0,330) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:39:35.772	Information	Transferring image block at (0,330) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:39:37.662	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:39:37.724	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:39:37.724	Information	[Instrument Response] 7 Progress 98% 10 seconds	GMT -5:00
12/19/23 12:39:39.677	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         142 V  	GMT -5:00
12/19/23 12:39:39.740	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:39:39.740	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:39:39.740	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:39:39.740	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:39:40.677	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          86 V  	GMT -5:00
12/19/23 12:39:40.755	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:39:41.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:39:42.692	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:39:43.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:39:43.754	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:39:45.363	Information	[Instrument Response] 7 Progress 100% 0 seconds	GMT -5:00
12/19/23 12:39:45.707	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:39:45.754	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         123 V  	GMT -5:00
12/19/23 12:39:45.754	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:39:45.754	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:39:45.754	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:39:46.707	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:39:47.363	Information	[Instrument Response] 7 ImageBlock 700 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:39:47.363	Information	[Instrument Response] 7 ImageBlock 800 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:39:47.363	Information	Transferring image block at (0,345) with size 290x10 to the 700 channel.	GMT -5:00
12/19/23 12:39:47.363	Information	[Instrument Response] 7 Status 0 COMPLETE SCAN 320.957 seconds	GMT -5:00
12/19/23 12:39:47.394	Information	Transferring image block at (0,345) with size 290x10 to the 800 channel.	GMT -5:00
12/19/23 12:39:47.410	Information	[Instrument Response] 7 Status 0 Averages (sec): setup=2.00005 line=0.378811 => total=271	GMT -5:00
12/19/23 12:39:47.457	Information	[Instrument Response] 7 Status 0 Comparisons: setup=1.00997 line=0.939575	GMT -5:00
12/19/23 12:39:47.457	Information	[Instrument Response] 7 Status 0 Finalize motors	GMT -5:00
12/19/23 12:39:47.457	Information	[Instrument Response] 7 Progress 100% Scan complete	GMT -5:00
12/19/23 12:39:47.472	Information	[Instrument Response] 7 Result 0 OK	GMT -5:00
12/19/23 12:39:47.472	Information	Acquisition 0000028_01 finished with channels: 700 800.	GMT -5:00
12/19/23 12:39:47.707	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 12:39:47.707	Information	Scan completed successfully	GMT -5:00
12/19/23 12:39:47.800	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/19/23 12:39:47.816	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          78 V  	GMT -5:00
12/19/23 12:39:47.832	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:39:47.832	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:39:47.832	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:39:48.718	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          75 V  	GMT -5:00
12/19/23 12:39:48.763	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          47 V  	GMT -5:00
12/19/23 12:39:49.723	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          46 V  	GMT -5:00
12/19/23 12:39:49.773	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          29 V  	GMT -5:00
12/19/23 12:39:49.773	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/19/23 12:39:50.728	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          30 V  	GMT -5:00
12/19/23 12:39:50.773	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          19 V  	GMT -5:00
12/19/23 12:39:51.731	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          20 V  	GMT -5:00
12/19/23 12:39:51.780	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          12 V  	GMT -5:00
12/19/23 12:39:52.734	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          13 V  	GMT -5:00
12/19/23 12:39:52.792	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/19/23 12:39:53.737	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 12:39:53.785	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 12:39:54.730	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/19/23 12:39:55.748	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/19/23 12:39:55.809	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/19/23 12:39:57.756	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/19/23 12:40:00.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:40:01.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:40:01.827	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/19/23 12:40:47.755	Information	[Instrument Request]  8 Stop 	GMT -5:00
12/19/23 12:40:47.802	Information	[Instrument Response] 8 Result 0 OK	GMT -5:00
12/19/23 12:40:48.333	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
12/19/23 12:41:38.870	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 12:41:38.870	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 12:41:38.870	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 12:41:38.870	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 12:41:38.948	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 12:41:38.979	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 12:41:38.995	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 12:41:38.995	Information	[Instrument Request]  9 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:41:41.291	Information	Starting acquisition...	GMT -5:00
12/19/23 12:41:42.119	Information	[Instrument Request]  10 ScanAuto 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:41:42.166	Information	[Instrument Response] 9 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Blocks=25	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:41:42.181	Information	[Instrument Response] 9 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 9 Result 0 OK	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.95 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Blocks=25	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:41:42.197	Information	[Instrument Response] 10 Progress 0% Initializing	GMT -5:00
12/19/23 12:41:42.213	Information	[Instrument Response] 10 Status 0 Preparation complete	GMT -5:00
12/19/23 12:41:45.134	Information	[Instrument Response] 10 Status 0 Motors initialized	GMT -5:00
12/19/23 12:41:45.181	Information	[Instrument Response] 10 Status 0 X-Motor configured	GMT -5:00
12/19/23 12:41:45.181	Information	[Instrument Response] 10 Status 0 DSP configured	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Cooling	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 355 xofs=1 yofs=0	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 355 xofs=0 yofs=0	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 start=34 scan_lines=15 line=0 end=355 700=Low,L1 800=High,4	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 start=34 scan_lines=30 line=0 end=355 700=High,4 800=Low,L1	GMT -5:00
12/19/23 12:41:45.228	Information	[Instrument Response] 10 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 12:41:45.962	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 12:41:46.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:41:46.024	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:41:49.852	Information	[Instrument Response] 10 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:41:49.977	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:41:50.039	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         163 V  	GMT -5:00
12/19/23 12:41:50.039	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:41:50.039	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:41:50.039	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:41:50.992	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:41:51.055	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:41:56.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:41:56.054	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/19/23 12:41:56.054	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:41:57.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:41:57.069	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:41:58.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:41:58.069	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/19/23 12:41:58.069	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:41:58.069	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:41:58.069	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:41:59.022	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:42:01.021	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:42:02.037	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:42:05.177	Information	[Instrument Response] 10 ImageBlock 700 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:05.177	Information	[Instrument Response] 10 ImageBlock 800 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:05.177	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:42:05.208	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:42:07.051	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/19/23 12:42:10.910	Information	[Instrument Response] 10 Progress 6% 5 minutes	GMT -5:00
12/19/23 12:42:11.082	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         136 V  	GMT -5:00
12/19/23 12:42:11.129	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:42:11.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:42:11.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:42:11.129	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:42:11.129	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 12:42:12.082	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:42:12.128	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:42:12.128	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:42:13.081	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:42:18.549	Information	[Instrument Response] 10 ImageBlock 700 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:18.549	Information	[Instrument Response] 10 ImageBlock 800 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:18.549	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:42:18.565	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:42:23.126	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:42:24.126	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:42:25.126	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:42:25.188	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:42:25.188	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:42:25.188	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:42:28.141	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:42:29.141	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:42:31.906	Information	[Instrument Response] 10 ImageBlock 700 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:31.906	Information	[Instrument Response] 10 ImageBlock 800 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:31.906	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:42:31.937	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:42:38.186	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         110 V  	GMT -5:00
12/19/23 12:42:38.248	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:42:38.248	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:42:38.248	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:42:39.185	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:42:41.154	Information	[Instrument Response] 10 Progress 15% 4 minutes, 30 seconds	GMT -5:00
12/19/23 12:42:45.278	Information	[Instrument Response] 10 ImageBlock 700 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:45.278	Information	[Instrument Response] 10 ImageBlock 800 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:45.278	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:42:45.309	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:42:50.230	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:42:51.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:42:51.292	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         131 V  	GMT -5:00
12/19/23 12:42:51.292	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:42:51.292	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:42:51.292	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:42:52.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/19/23 12:42:53.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:42:58.635	Information	[Instrument Response] 10 ImageBlock 700 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:58.635	Information	[Instrument Response] 10 ImageBlock 800 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:42:58.650	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:42:58.666	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:43:05.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:43:05.352	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:43:05.352	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:43:05.352	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:43:06.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:43:10.913	Information	[Instrument Response] 10 Progress 25% 4 minutes	GMT -5:00
12/19/23 12:43:12.007	Information	[Instrument Response] 10 ImageBlock 700 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:12.007	Information	[Instrument Response] 10 ImageBlock 800 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:12.007	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:43:12.038	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:43:17.349	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:43:17.396	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:43:18.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:43:18.412	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         107 V  	GMT -5:00
12/19/23 12:43:18.412	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:43:18.412	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:43:18.412	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:43:18.412	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 12:43:19.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:43:22.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:43:22.427	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:43:23.379	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:43:23.426	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:43:25.363	Information	[Instrument Response] 10 ImageBlock 700 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:25.363	Information	[Instrument Response] 10 ImageBlock 800 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:25.379	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:43:25.395	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:43:25.457	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:43:26.379	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:43:31.409	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         124 V  	GMT -5:00
12/19/23 12:43:31.472	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:43:31.472	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:43:31.472	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:43:32.425	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:43:38.736	Information	[Instrument Response] 10 ImageBlock 700 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:38.736	Information	[Instrument Response] 10 ImageBlock 800 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:38.736	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:43:38.767	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:43:41.048	Information	[Instrument Response] 10 Progress 35% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:43:44.469	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         162 V  	GMT -5:00
12/19/23 12:43:44.547	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:43:44.547	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:43:44.547	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:43:44.547	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:43:45.484	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:43:45.531	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:43:45.531	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/19/23 12:43:45.531	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:43:46.484	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:43:52.108	Information	[Instrument Response] 10 ImageBlock 700 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:52.108	Information	[Instrument Response] 10 ImageBlock 800 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:43:52.108	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:43:52.124	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:43:52.514	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:43:52.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:43:52.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:43:53.514	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:43:53.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:43:53.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:43:58.529	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/19/23 12:43:58.607	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:43:58.607	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:43:58.607	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:43:59.544	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:44:05.465	Information	[Instrument Response] 10 ImageBlock 700 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:05.465	Information	[Instrument Response] 10 ImageBlock 800 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:05.465	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:44:05.496	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:44:11.198	Information	[Instrument Response] 10 Progress 44% 3 minutes	GMT -5:00
12/19/23 12:44:11.588	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:44:11.651	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/19/23 12:44:11.651	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:44:11.651	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:44:12.604	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:44:18.837	Information	[Instrument Response] 10 ImageBlock 700 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:18.837	Information	[Instrument Response] 10 ImageBlock 800 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:18.837	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:44:18.852	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:44:19.634	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:44:20.633	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:44:22.633	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:44:23.648	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:44:24.648	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         144 V  	GMT -5:00
12/19/23 12:44:24.695	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:44:24.695	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:44:24.695	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:44:24.695	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:44:25.648	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/19/23 12:44:25.710	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:44:25.710	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:44:26.648	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:44:27.663	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:44:28.663	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:44:32.194	Information	[Instrument Response] 10 ImageBlock 700 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:32.194	Information	[Instrument Response] 10 ImageBlock 800 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:32.194	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:44:32.225	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:44:38.708	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:44:38.770	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/19/23 12:44:38.770	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:44:38.770	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:44:39.723	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:44:41.051	Information	[Instrument Response] 10 Progress 53% 2 minutes, 30 seconds	GMT -5:00
12/19/23 12:44:45.566	Information	[Instrument Response] 10 ImageBlock 700 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:45.566	Information	[Instrument Response] 10 ImageBlock 800 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:45.566	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:44:45.597	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:44:51.768	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/19/23 12:44:51.830	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:44:51.830	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:44:51.830	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:44:52.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:44:54.783	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:44:55.782	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:44:58.922	Information	[Instrument Response] 10 ImageBlock 700 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:58.922	Information	[Instrument Response] 10 ImageBlock 800 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:44:58.922	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:44:58.954	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:45:04.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:45:04.890	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         136 V  	GMT -5:00
12/19/23 12:45:04.890	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:45:04.890	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:45:04.890	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:45:05.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:45:05.890	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:45:06.843	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:45:11.201	Information	[Instrument Response] 10 Progress 63% 2 minutes	GMT -5:00
12/19/23 12:45:12.295	Information	[Instrument Response] 10 ImageBlock 700 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:12.295	Information	[Instrument Response] 10 ImageBlock 800 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:12.295	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:45:12.326	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:45:16.872	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:45:16.934	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:45:17.887	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:45:17.934	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:45:18.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:45:18.934	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:45:18.934	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:45:18.934	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:45:21.152	Information	[Instrument Response] 10 Progress 66% 1 minute, 50 seconds	GMT -5:00
12/19/23 12:45:24.901	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:45:24.964	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:45:25.651	Information	[Instrument Response] 10 ImageBlock 700 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:25.651	Information	[Instrument Response] 10 ImageBlock 800 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:25.667	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:45:25.683	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:45:25.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:45:25.979	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:45:30.994	Information	[Instrument Response] 10 Progress 70% 1 minute, 40 seconds	GMT -5:00
12/19/23 12:45:31.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:45:32.009	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         110 V  	GMT -5:00
12/19/23 12:45:32.009	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:45:32.009	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:45:32.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:45:39.024	Information	[Instrument Response] 10 ImageBlock 700 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:39.024	Information	[Instrument Response] 10 ImageBlock 800 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:39.024	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:45:39.055	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:45:40.961	Information	[Instrument Response] 10 Progress 73% 1 minute, 30 seconds	GMT -5:00
12/19/23 12:45:45.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 12:45:45.054	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:45:45.054	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:45:45.054	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:45:45.054	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:45:46.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:45:46.053	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:45:47.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:45:50.912	Information	[Instrument Response] 10 Progress 75% 1 minute, 20 seconds	GMT -5:00
12/19/23 12:45:52.021	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:45:52.083	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:45:52.380	Information	[Instrument Response] 10 ImageBlock 700 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:52.380	Information	[Instrument Response] 10 ImageBlock 800 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:45:52.396	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:45:52.411	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:45:53.036	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:45:53.099	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:45:59.066	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:45:59.113	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:45:59.113	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:45:59.113	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:46:00.066	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:46:00.863	Information	[Instrument Response] 10 Progress 78% 1 minute, 10 seconds	GMT -5:00
12/19/23 12:46:05.752	Information	[Instrument Response] 10 ImageBlock 700 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:05.752	Information	[Instrument Response] 10 ImageBlock 800 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:05.752	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:46:05.784	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:46:11.095	Information	[Instrument Response] 10 Progress 82% 1 minute	GMT -5:00
12/19/23 12:46:12.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:46:12.173	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:46:12.173	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:46:12.173	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:46:13.126	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:46:19.125	Information	[Instrument Response] 10 ImageBlock 700 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:19.125	Information	[Instrument Response] 10 ImageBlock 800 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:19.125	Information	Transferring image block at (0,285) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:46:19.140	Information	Transferring image block at (0,285) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:46:19.203	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:46:19.203	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:46:20.140	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:46:20.203	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:46:21.046	Information	[Instrument Response] 10 Progress 85% 50 seconds	GMT -5:00
12/19/23 12:46:25.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:46:25.233	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/19/23 12:46:25.233	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:46:25.233	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:46:26.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:46:27.186	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:46:31.013	Information	[Instrument Response] 10 Progress 88% 40 seconds	GMT -5:00
12/19/23 12:46:32.481	Information	[Instrument Response] 10 ImageBlock 700 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:32.481	Information	[Instrument Response] 10 ImageBlock 800 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:32.481	Information	Transferring image block at (0,300) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:46:32.513	Information	Transferring image block at (0,300) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:46:38.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         150 V  	GMT -5:00
12/19/23 12:46:38.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:46:38.277	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         161 V  	GMT -5:00
12/19/23 12:46:38.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:46:38.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:46:39.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/19/23 12:46:39.292	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:46:39.292	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:46:40.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:46:40.292	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:46:40.964	Information	[Instrument Response] 10 Progress 91% 30 seconds	GMT -5:00
12/19/23 12:46:45.854	Information	[Instrument Response] 10 ImageBlock 700 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:45.854	Information	[Instrument Response] 10 ImageBlock 800 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:45.854	Information	Transferring image block at (0,315) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:46:45.869	Information	Transferring image block at (0,315) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:46:46.260	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:46:47.259	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:46:49.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:46:50.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:46:50.321	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:46:51.196	Information	[Instrument Response] 10 Progress 95% 20 seconds	GMT -5:00
12/19/23 12:46:51.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:46:52.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:46:52.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:46:52.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         103 V  	GMT -5:00
12/19/23 12:46:52.337	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:46:52.337	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:46:53.290	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:46:53.336	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:46:59.210	Information	[Instrument Response] 10 ImageBlock 700 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:59.210	Information	[Instrument Response] 10 ImageBlock 800 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:46:59.210	Information	Transferring image block at (0,330) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:46:59.241	Information	Transferring image block at (0,330) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:47:01.147	Information	[Instrument Response] 10 Progress 98% 10 seconds	GMT -5:00
12/19/23 12:47:03.334	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/19/23 12:47:03.397	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 12:47:03.397	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:47:03.397	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:47:03.397	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:47:04.334	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:47:08.833	Information	[Instrument Response] 10 Progress 100% 0 seconds	GMT -5:00
12/19/23 12:47:09.364	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:47:09.411	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/19/23 12:47:09.411	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:47:09.411	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:47:10.364	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:47:10.833	Information	[Instrument Response] 10 ImageBlock 700 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:47:10.833	Information	[Instrument Response] 10 ImageBlock 800 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:47:10.833	Information	[Instrument Response] 10 Status 0 COMPLETE SCAN 320.961 seconds	GMT -5:00
12/19/23 12:47:10.833	Information	Transferring image block at (0,345) with size 290x10 to the 700 channel.	GMT -5:00
12/19/23 12:47:10.833	Information	[Instrument Response] 10 Status 0 Averages (sec): setup=2.00005 line=0.378816 => total=271	GMT -5:00
12/19/23 12:47:10.864	Information	Transferring image block at (0,345) with size 290x10 to the 800 channel.	GMT -5:00
12/19/23 12:47:10.864	Information	[Instrument Response] 10 Status 0 Comparisons: setup=1.00997 line=0.939561	GMT -5:00
12/19/23 12:47:10.926	Information	[Instrument Response] 10 Status 0 Finalize motors	GMT -5:00
12/19/23 12:47:10.926	Information	[Instrument Response] 10 Progress 100% Scan complete	GMT -5:00
12/19/23 12:47:10.942	Information	[Instrument Response] 10 Result 0 OK	GMT -5:00
12/19/23 12:47:10.942	Information	Acquisition 0000029_01 finished with channels: 700 800.	GMT -5:00
12/19/23 12:47:11.145	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 12:47:11.145	Information	Scan completed successfully	GMT -5:00
12/19/23 12:47:11.364	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/19/23 12:47:11.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 12:47:11.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          71 V  	GMT -5:00
12/19/23 12:47:11.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:47:11.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:47:11.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:47:12.377	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          68 V  	GMT -5:00
12/19/23 12:47:12.421	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          43 V  	GMT -5:00
12/19/23 12:47:13.376	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          42 V  	GMT -5:00
12/19/23 12:47:13.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          26 V  	GMT -5:00
12/19/23 12:47:13.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/19/23 12:47:14.376	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          28 V  	GMT -5:00
12/19/23 12:47:14.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          18 V  	GMT -5:00
12/19/23 12:47:15.389	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          19 V  	GMT -5:00
12/19/23 12:47:15.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          12 V  	GMT -5:00
12/19/23 12:47:16.392	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/19/23 12:47:16.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/19/23 12:47:17.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 12:47:17.445	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 12:47:18.401	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/19/23 12:47:18.455	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/19/23 12:47:20.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/19/23 12:47:21.418	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/19/23 12:47:23.408	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/19/23 12:47:25.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:47:35.049	Information	[Instrument Request]  11 Stop 	GMT -5:00
12/19/23 12:47:35.096	Information	[Instrument Response] 11 Result 0 OK	GMT -5:00
12/19/23 12:47:36.002	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
12/19/23 12:48:12.182	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 12:48:12.182	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 12:48:12.182	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 12:48:12.182	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 12:48:12.260	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 12:48:12.291	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 12:48:12.307	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 12:48:12.307	Information	[Instrument Request]  12 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Blocks=25	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:48:15.259	Information	[Instrument Response] 12 Result 0 OK	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.98 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/19/23 12:48:15.275	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/19/23 12:48:15.322	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 12:48:19.649	Information	Starting acquisition...	GMT -5:00
12/19/23 12:48:20.446	Information	[Instrument Request]  13 ScanAuto 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 Computed scan parameters	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 12:48:20.508	Information	[Instrument Response] 13 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 12:48:20.524	Information	[Instrument Response] 13 Status 0 Blocks=25	GMT -5:00
12/19/23 12:48:20.524	Information	[Instrument Response] 13 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 12:48:20.524	Information	[Instrument Response] 13 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:48:20.524	Information	[Instrument Response] 13 Progress 0% Initializing	GMT -5:00
12/19/23 12:48:20.540	Information	[Instrument Response] 13 Status 0 Preparation complete	GMT -5:00
12/19/23 12:48:23.461	Information	[Instrument Response] 13 Status 0 Motors initialized	GMT -5:00
12/19/23 12:48:23.523	Information	[Instrument Response] 13 Status 0 X-Motor configured	GMT -5:00
12/19/23 12:48:23.523	Information	[Instrument Response] 13 Status 0 DSP configured	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Cooling	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 355 xofs=1 yofs=0	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 355 xofs=0 yofs=0	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 start=34 scan_lines=15 line=0 end=355 700=Low,L1 800=High,4	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 start=34 scan_lines=30 line=0 end=355 700=High,4 800=Low,L1	GMT -5:00
12/19/23 12:48:23.570	Information	[Instrument Response] 13 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 12:48:24.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 12:48:24.351	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:48:28.194	Information	[Instrument Response] 13 Progress 0% 6 minutes	GMT -5:00
12/19/23 12:48:28.319	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:48:28.366	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         163 V  	GMT -5:00
12/19/23 12:48:28.366	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:48:28.366	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:48:28.366	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:48:29.319	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:48:29.382	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:48:31.334	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:48:32.334	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:48:34.334	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 12:48:36.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:48:36.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/19/23 12:48:36.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:48:36.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:48:37.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:48:39.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:48:40.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:48:43.519	Information	[Instrument Response] 13 ImageBlock 700 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:48:43.519	Information	[Instrument Response] 13 ImageBlock 800 0 0 290 15 290 355 8700	GMT -5:00
12/19/23 12:48:43.519	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:48:43.551	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:48:49.237	Information	[Instrument Response] 13 Progress 6% 5 minutes	GMT -5:00
12/19/23 12:48:49.393	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/19/23 12:48:49.456	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:48:49.456	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:48:49.456	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:48:49.456	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:48:50.409	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:48:50.455	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:48:51.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:48:56.876	Information	[Instrument Response] 13 ImageBlock 700 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:48:56.876	Information	[Instrument Response] 13 ImageBlock 800 0 15 290 15 290 355 8700	GMT -5:00
12/19/23 12:48:56.876	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:48:56.907	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:48:58.438	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      315 mA 	GMT -5:00
12/19/23 12:48:58.485	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:48:59.438	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:49:01.453	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:49:01.500	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:49:02.453	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:49:02.500	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:49:03.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:49:03.515	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:49:03.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:49:03.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:49:06.468	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:49:07.468	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:49:10.248	Information	[Instrument Response] 13 ImageBlock 700 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:10.248	Information	[Instrument Response] 13 ImageBlock 800 0 30 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:10.248	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:49:10.279	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:49:16.513	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         110 V  	GMT -5:00
12/19/23 12:49:16.575	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:49:16.575	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:49:16.575	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:49:17.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:49:19.481	Information	[Instrument Response] 13 Progress 15% 4 minutes, 30 seconds	GMT -5:00
12/19/23 12:49:23.605	Information	[Instrument Response] 13 ImageBlock 700 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:23.605	Information	[Instrument Response] 13 ImageBlock 800 0 45 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:23.620	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:49:23.636	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:49:29.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:49:29.619	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         131 V  	GMT -5:00
12/19/23 12:49:29.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:49:29.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:49:29.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:49:30.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/19/23 12:49:30.635	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:49:31.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:49:33.587	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:49:34.587	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:49:36.977	Information	[Instrument Response] 13 ImageBlock 700 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:36.977	Information	[Instrument Response] 13 ImageBlock 800 0 60 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:36.977	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:49:37.008	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:49:43.632	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:49:43.694	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:49:43.694	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:49:43.694	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:49:44.647	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:49:45.647	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 12:49:49.240	Information	[Instrument Response] 13 Progress 25% 4 minutes	GMT -5:00
12/19/23 12:49:50.334	Information	[Instrument Response] 13 ImageBlock 700 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:50.349	Information	[Instrument Response] 13 ImageBlock 800 0 75 290 15 290 355 8700	GMT -5:00
12/19/23 12:49:50.349	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:49:50.365	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:49:56.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:49:56.739	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         107 V  	GMT -5:00
12/19/23 12:49:56.739	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:49:56.739	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:49:57.692	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:50:00.707	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:50:01.706	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:50:03.706	Information	[Instrument Response] 13 ImageBlock 700 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:03.706	Information	[Instrument Response] 13 ImageBlock 800 0 90 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:03.706	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:50:03.737	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:50:03.784	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:50:04.721	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:50:09.752	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         124 V  	GMT -5:00
12/19/23 12:50:09.814	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:50:09.814	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:50:09.814	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:50:10.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:50:17.078	Information	[Instrument Response] 13 ImageBlock 700 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:17.078	Information	[Instrument Response] 13 ImageBlock 800 0 105 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:17.078	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:50:17.094	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:50:19.390	Information	[Instrument Response] 13 Progress 35% 3 minutes, 30 seconds	GMT -5:00
12/19/23 12:50:22.811	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         159 V  	GMT -5:00
12/19/23 12:50:22.874	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         151 V  	GMT -5:00
12/19/23 12:50:22.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:50:22.874	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          35 mA 	GMT -5:00
12/19/23 12:50:22.874	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:50:23.811	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:50:23.874	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/19/23 12:50:23.874	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:50:24.827	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:50:30.435	Information	[Instrument Response] 13 ImageBlock 700 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:30.435	Information	[Instrument Response] 13 ImageBlock 800 0 120 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:30.435	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:50:30.466	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:50:30.841	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:50:30.903	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 12:50:31.856	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:50:31.903	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 12:50:33.856	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:50:34.856	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:50:36.871	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/19/23 12:50:36.918	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:50:36.918	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:50:36.918	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:50:37.871	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:50:43.807	Information	[Instrument Response] 13 ImageBlock 700 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:43.807	Information	[Instrument Response] 13 ImageBlock 800 0 135 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:43.807	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:50:43.838	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:50:49.525	Information	[Instrument Response] 13 Progress 44% 3 minutes	GMT -5:00
12/19/23 12:50:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:50:49.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/19/23 12:50:49.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:50:49.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:50:50.931	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:50:57.164	Information	[Instrument Response] 13 ImageBlock 700 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:57.164	Information	[Instrument Response] 13 ImageBlock 800 0 150 290 15 290 355 8700	GMT -5:00
12/19/23 12:50:57.164	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:50:57.195	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:51:00.975	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:51:01.975	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:51:02.991	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         144 V  	GMT -5:00
12/19/23 12:51:03.037	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:51:03.037	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:51:03.037	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:51:03.037	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:51:03.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/19/23 12:51:04.037	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:51:04.037	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:51:04.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:51:10.536	Information	[Instrument Response] 13 ImageBlock 700 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:10.536	Information	[Instrument Response] 13 ImageBlock 800 0 165 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:10.536	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:51:10.567	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:51:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:51:17.113	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/19/23 12:51:17.113	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:51:17.113	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:51:18.050	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:51:19.393	Information	[Instrument Response] 13 Progress 53% 2 minutes, 30 seconds	GMT -5:00
12/19/23 12:51:23.892	Information	[Instrument Response] 13 ImageBlock 700 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:23.892	Information	[Instrument Response] 13 ImageBlock 800 0 180 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:23.908	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:51:23.924	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:51:28.095	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:51:29.095	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:51:30.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/19/23 12:51:30.157	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:51:30.157	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:51:30.157	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:51:31.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:51:33.109	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/19/23 12:51:34.125	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:51:37.265	Information	[Instrument Response] 13 ImageBlock 700 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:37.265	Information	[Instrument Response] 13 ImageBlock 800 0 195 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:37.265	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:51:37.296	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:51:43.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 12:51:43.217	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         136 V  	GMT -5:00
12/19/23 12:51:43.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:51:43.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:51:43.217	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:51:44.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 12:51:44.216	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:51:45.169	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:51:49.528	Information	[Instrument Response] 13 Progress 63% 2 minutes	GMT -5:00
12/19/23 12:51:50.621	Information	[Instrument Response] 13 ImageBlock 700 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:50.621	Information	[Instrument Response] 13 ImageBlock 800 0 210 290 15 290 355 8700	GMT -5:00
12/19/23 12:51:50.637	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:51:50.653	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:51:55.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/19/23 12:51:56.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:51:57.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:51:57.276	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:51:57.276	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:51:57.276	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:51:59.495	Information	[Instrument Response] 13 Progress 66% 1 minute, 50 seconds	GMT -5:00
12/19/23 12:52:00.229	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:52:01.229	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:52:03.994	Information	[Instrument Response] 13 ImageBlock 700 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:03.994	Information	[Instrument Response] 13 ImageBlock 800 0 225 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:03.994	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:52:04.025	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:52:09.336	Information	[Instrument Response] 13 Progress 70% 1 minute, 40 seconds	GMT -5:00
12/19/23 12:52:10.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:52:10.336	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/19/23 12:52:10.336	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:52:10.336	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:52:11.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:52:17.366	Information	[Instrument Response] 13 ImageBlock 700 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:17.366	Information	[Instrument Response] 13 ImageBlock 800 0 240 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:17.366	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:52:17.381	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:52:19.287	Information	[Instrument Response] 13 Progress 73% 1 minute, 30 seconds	GMT -5:00
12/19/23 12:52:23.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 12:52:23.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:52:23.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:52:23.396	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:52:24.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 12:52:25.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:52:27.348	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:52:28.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:52:29.254	Information	[Instrument Response] 13 Progress 75% 1 minute, 20 seconds	GMT -5:00
12/19/23 12:52:30.722	Information	[Instrument Response] 13 ImageBlock 700 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:30.722	Information	[Instrument Response] 13 ImageBlock 800 0 255 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:30.722	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:52:30.754	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:52:37.409	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:52:37.455	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 12:52:37.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:52:37.455	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:52:38.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:52:39.205	Information	[Instrument Response] 13 Progress 78% 1 minute, 10 seconds	GMT -5:00
12/19/23 12:52:44.095	Information	[Instrument Response] 13 ImageBlock 700 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:44.095	Information	[Instrument Response] 13 ImageBlock 800 0 270 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:44.095	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:52:44.110	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:52:49.437	Information	[Instrument Response] 13 Progress 82% 1 minute	GMT -5:00
12/19/23 12:52:50.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 12:52:50.515	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:52:50.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:52:50.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:52:51.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:52:57.451	Information	[Instrument Response] 13 ImageBlock 700 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:57.451	Information	[Instrument Response] 13 ImageBlock 800 0 285 290 15 290 355 8700	GMT -5:00
12/19/23 12:52:57.451	Information	Transferring image block at (0,285) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:52:57.483	Information	Transferring image block at (0,285) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:52:57.529	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 12:52:58.482	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:52:59.388	Information	[Instrument Response] 13 Progress 85% 50 seconds	GMT -5:00
12/19/23 12:53:00.498	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:53:01.497	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:53:03.513	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:53:03.559	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/19/23 12:53:03.559	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:53:03.559	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:53:04.512	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:53:05.512	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:53:09.339	Information	[Instrument Response] 13 Progress 88% 40 seconds	GMT -5:00
12/19/23 12:53:10.824	Information	[Instrument Response] 13 ImageBlock 700 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:10.824	Information	[Instrument Response] 13 ImageBlock 800 0 300 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:10.824	Information	Transferring image block at (0,300) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:53:10.855	Information	Transferring image block at (0,300) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:53:16.557	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         150 V  	GMT -5:00
12/19/23 12:53:16.619	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         158 V  	GMT -5:00
12/19/23 12:53:16.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 12:53:16.619	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:53:16.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:53:17.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/19/23 12:53:17.619	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 12:53:17.619	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:53:18.572	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:53:19.306	Information	[Instrument Response] 13 Progress 91% 30 seconds	GMT -5:00
12/19/23 12:53:24.180	Information	[Instrument Response] 13 ImageBlock 700 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:24.180	Information	[Instrument Response] 13 ImageBlock 800 0 315 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:24.180	Information	Transferring image block at (0,315) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:53:24.211	Information	Transferring image block at (0,315) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:53:24.602	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 12:53:25.602	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:53:27.601	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/19/23 12:53:27.664	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 12:53:28.617	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 12:53:29.523	Information	[Instrument Response] 13 Progress 95% 20 seconds	GMT -5:00
12/19/23 12:53:30.616	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:53:30.679	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         103 V  	GMT -5:00
12/19/23 12:53:30.679	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:53:30.679	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:53:31.616	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 12:53:31.679	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 12:53:37.552	Information	[Instrument Response] 13 ImageBlock 700 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:37.552	Information	[Instrument Response] 13 ImageBlock 800 0 330 290 15 290 355 8700	GMT -5:00
12/19/23 12:53:37.552	Information	Transferring image block at (0,330) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 12:53:37.584	Information	Transferring image block at (0,330) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 12:53:39.489	Information	[Instrument Response] 13 Progress 98% 10 seconds	GMT -5:00
12/19/23 12:53:41.676	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/19/23 12:53:41.723	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:53:41.723	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 12:53:41.723	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 12:53:42.676	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 12:53:47.160	Information	[Instrument Response] 13 Progress 100% 0 seconds	GMT -5:00
12/19/23 12:53:47.691	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 12:53:47.753	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/19/23 12:53:47.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 12:53:47.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 12:53:48.691	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 12:53:49.159	Information	[Instrument Response] 13 ImageBlock 700 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:53:49.159	Information	[Instrument Response] 13 ImageBlock 800 0 345 290 10 290 355 5800	GMT -5:00
12/19/23 12:53:49.159	Information	[Instrument Response] 13 Status 0 COMPLETE SCAN 320.961 seconds	GMT -5:00
12/19/23 12:53:49.175	Information	Transferring image block at (0,345) with size 290x10 to the 700 channel.	GMT -5:00
12/19/23 12:53:49.175	Information	[Instrument Response] 13 Status 0 Averages (sec): setup=2.00005 line=0.378816 => total=271	GMT -5:00
12/19/23 12:53:49.191	Information	Transferring image block at (0,345) with size 290x10 to the 800 channel.	GMT -5:00
12/19/23 12:53:49.191	Information	[Instrument Response] 13 Status 0 Comparisons: setup=1.00997 line=0.939561	GMT -5:00
12/19/23 12:53:49.253	Information	[Instrument Response] 13 Status 0 Finalize motors	GMT -5:00
12/19/23 12:53:49.253	Information	[Instrument Response] 13 Progress 100% Scan complete	GMT -5:00
12/19/23 12:53:49.284	Information	[Instrument Response] 13 Result 0 OK	GMT -5:00
12/19/23 12:53:49.284	Information	Acquisition 0000030_01 finished with channels: 700 800.	GMT -5:00
12/19/23 12:53:49.550	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 12:53:49.550	Information	Scan completed successfully	GMT -5:00
12/19/23 12:53:49.706	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/19/23 12:53:49.837	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          71 V  	GMT -5:00
12/19/23 12:53:49.837	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 12:53:49.837	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 12:53:49.837	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 12:53:50.710	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          68 V  	GMT -5:00
12/19/23 12:53:50.755	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          43 V  	GMT -5:00
12/19/23 12:53:51.715	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          42 V  	GMT -5:00
12/19/23 12:53:51.761	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          26 V  	GMT -5:00
12/19/23 12:53:52.721	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          28 V  	GMT -5:00
12/19/23 12:53:52.761	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          18 V  	GMT -5:00
12/19/23 12:53:53.723	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          19 V  	GMT -5:00
12/19/23 12:53:53.772	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/19/23 12:53:54.726	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/19/23 12:53:54.785	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/19/23 12:53:55.728	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 12:53:55.776	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 12:53:56.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/19/23 12:53:56.795	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/19/23 12:53:57.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/19/23 12:53:59.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/19/23 12:53:59.801	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 12:54:00.741	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/19/23 12:54:42.332	Information	Successfully saved C:\Users\wmao2\2023-12-19\ACTR1.tif	GMT -5:00
12/19/23 12:54:48.330	Information	[Instrument Request]  14 Stop 	GMT -5:00
12/19/23 12:54:48.330	Information	[Instrument Response] 14 Result 0 OK	GMT -5:00
12/19/23 12:54:49.330	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
